#### Correction Factor Tables for Four-Point Probe Resistivity Measurements on Thin, Circular Semiconductor Samples

**Question: What is the need for correction factors in four point probe measurements?**

**Answer:** The geometry of the sample determines the correction factors that must be used, additionally the position of the probes on the sample and the spacings between the probes. The need for correction factors is caused by the proximity of a boundary which limits the possible current paths in the sample. The most basic sample would be semi-infinite in extent i.e., it extends to infinity in all directions below the plane in which the four probes are located. All other cases would restrict the current paths available, eg., an infinite plane sample of finite thickness requires a correction factor based on the thickness.

**The table shown immediately below pertains to one of the most commonly needed correction factors, i.e., correction factors for measuring a “thin, circular slice”. After the table, you will find links to all of the pages from the 1964 National Bureau of Standards Technical Note 199, “Correction Factor Tables for Four-Point Probe Resistivity Measurements on Thin, Circular Semiconductor Samples”, with each page available as an individual PDF file. The document provides extensive tables of geometrical correction factors. **

**Correction factors for measuring a thin circular slice, measured in the center**

The following table pertains to sheet resistance measurements made in the center of a circular slice. **d/s **= **d**iameter of sample divided by probe **s**pacing (probe spacing being the distance between any two adjacent probes). For example, a 4mm diameter sample probed with a four point probe with 1mm tip spacing would have a correction factor of 0.6462. A 100mm wafer measured with a four point probe head that has 1mm tip spacing would have a correction factor of 0.9991. A result better than 0.1% can be obtained by measuring in the center of a circle with diameter greater than 100 x s., better than 1% is obtained with 40 x s.

**National Bureau of Standards Technical Note 199 – “Correction Factor Tables for Four-Point Probe Resistivity Measurements on Thin, Circular Semiconductor Samples**“

**nbs.pdf** 6.08 MEG PDF file which includes all 42 pages of the 1964 National Bureau of Standards Technical Note 199, *“Correction Factor Tables for Four-Point Probe Resistivity Measurements on Thin, Circular Semiconductor Samples”*

The individual pages range in size from 60K to 230K.

The individual pages:

nbs1.pdf NBS Document Cover Page

nbs2.pdf NBS Document Introduction Page I

nbs3.pdf NBS Document Introduction Page II

nbs4.pdf NBS Document Introduction Page III

nbs5.pdf NBS Document Introduction Page IV

nbs6.pdf NBS Document Introduction Page V

nbs7.pdf NBS Document References Page VI

nbs8.pdf NBS Document Geometry Page VII

nbs9.pdf NBS Document Tables, page 1

nbs10.pdf NBS Document Tables, page 2

nbs11.pdf NBS Document Tables, page 3

nbs12.pdf NBS Document Tables, page 4

nbs13.pdf NBS Document Tables, page 5

nbs14.pdf NBS Document Tables, page 6

nbs15.pdf NBS Document Tables, page 7

nbs16.pdf NBS Document Tables, page 8

nbs17.pdf NBS Document Tables, page 9

nbs18.pdf NBS Document Tables, page 10

nbs19.pdf NBS Document Tables, page 11

nbs20.pdf NBS Document Tables, page 12

nbs21.pdf NBS Document Tables, page 13

nbs22.pdf NBS Document Tables, page 14

nbs23.pdf NBS Document Tables, page 15

nbs24.pdf NBS Document Tables, page 16

nbs25.pdf NBS Document Tables, page 17

nbs26.pdf NBS Document Tables, page 18

nbs27.pdf NBS Document Tables, page 19

nbs28.pdf NBS Document Tables, page 20

nbs29.pdf NBS Document Tables, page 21

nbs30.pdf NBS Document Tables, page 22

nbs31.pdf NBS Document Tables, page 23

nbs32.pdf NBS Document Tables, page 24

nbs33.pdf NBS Document Tables, page 25

nbs34.pdf NBS Document Tables, page 26

nbs35.pdf NBS Document Tables, page 27

nbs36.pdf NBS Document Tables, page 28

nbs37.pdf NBS Document Tables, page 29

nbs38.pdf NBS Document Tables, page 30

nbs39.pdf NBS Document Tables, page 31

nbs40.pdf NBS Document Tables, page 32

nbs41.pdf NBS Document Tables, page 33

nbs42.pdf NBS Document Tables, page 34

**Four-Point-Probes** is a division of Bridge Technology. To request further information please call Bridge Technology at (480) 219-9007 or send e-mail to Joshua Bridge at: sales@bridgetec.com